Tips about the vacorda glass plate level gauge
Author:vacorda Source:vacorda.com Date: 2018-04-11 15:19:15
Working condition and structure
The glass plate type liquid level meter is designed based on the principle of the communicator. The liquid passage formed by the glass plate and the liquid level gauge body is connected to the measured container through a flange or a taper pipe thread to form a communicator, which is observed through the glass plate. The liquid level is the same as the liquid level in the container, that is, the liquid level.
The needle valve at both ends of the level gauge not only functions as a stop valve, but also has the function of a check valve in the internal steel ball. When the liquid level gauge accidentally breaks and leaks, the steel ball can automatically close the liquid passage under the action of the medium pressure. To prevent the large outflow of liquid for safety protection. The level gauge can change the material of the part or add some ancillary parts to achieve anti-corrosion, thermal insulation, anti-frost, lighting and other functions. Widely used in petroleum, chemical industry, electric power, metallurgy, medicine and other fields.
▲The reading is clear, intuitive and reliable.
▲ simple structure, easy maintenance.
▲Durable, it is an ideal instrument for observation of the liquid level in various medium and low pressure vessels.
Tips for installation
1. During installation, in order to ensure the function of automatic sealing, the liquid in the container must have a certain pressure, at least more than 0.2MPa. When the upper and lower valves are opened, the valve stem quitting cannot be less than four turns. When the steel ball seals the door, it does not hit the top of the valve stem to avoid damage.
2. In the process of transportation and unpacking, we must be careful not to encounter something harder to avoid breaking the glass plate.
3. Also pay attention to the specifications and models, cannot install the wrong gauge level gauge, some media on the leveler is limited, such as some of the glass or steel plate corrosive media cannot be used.
4. Other maintenance clean, wash and install with the glass tube level gauge is similar.
Sichuan Vacorda Instruments Manufacturing Co., Ltd
21 Years Focused on Level Measurement in Extreme Process Conditions
Tel: +0086 28 8701 3699