1.Medium has changed
2.Impurity has stucked the float ball, and we need to clean up timely.
1.Check whether there is block in by pass pipe.
2.Check the the density of medium.
3.Check whether the float ball is stucked by float ball.
The length is customized, but in order to be convenient for transport, if the length is over 6m, we recommend.
Reading level will be higher than true level, usually higher length the diameter of float ball minus 70mm.
1. As for unconversible switch alarm, we just need to directly connect amphenol connector;
2. As for conversible swich alarm, we could connect with NO-COM or NC-COM accordingly.
1.Check work condition of flapper part of level gauge or check work condition of float ball.
2.If the local part is ok, we should check if there is breakdown in transmitter.
1.Before installation, check specification with label plate.
2.There shouldn't magnetic field within 400m.
1.Check working power supply;
3.Check load resistance;
4.Cerify measuring range, zero point.
Medium density,working temperature, working pressure, material, flange center to center distance, flange standard, whether need 4-20mA output signal; whether need level alarm.
They are same at working principle and structure. But magnetic flap liquid level gauge's working temperature is higher.
Not necessary. it depends on case.
The flapper are stucked.
The structure and material of float ball depend on medium density, working pressure, working temperature and corrosivity.
It could be, but we need to fabricate in sections to be convenient for transport.
Not necessary, it could be customized.
It can't be changed. the position of flange depends on flange center to center distance.
The lowest density is 0.45g/cm3.
Yes, we have special type of high temperature, and highest temperature could be 500℃.
We could use anti-frost magnetic flap liquid level gauge.
We recommend you the magnetic flap liquid level gauge.
Sure, material of float ball should be titanium.
It's safe. If medium conform to requirements, the level gauge is very reliable.
DN means diameter of pipe.