How much is the magnetic flap level gauge? This question is difficult to answer uniformly, because the material of the magnetic flap level gauge is different, and the price will be different. As a professional manufacturer of magnetic flap level gauges, Huikeda Instruments will answer their questions about magnetic flap level gauges.How much is the magnetic flap level gaugeThe magnetic flap level gauge is a customized instrument product. The main factors that determine its price are scope, material and flange. The price of standard 304 magnetic flap level gauge on the market is about 800-900 yuan, the price of 316 magnetic flap level gauge is about 1200-1500 yuan, and the price of stainless steel lined PTFE anticorrosive magnetic flap level gauge is around 800-900 yuan. About 1500-2000 yuan, of course, subject to the restrictions of the above-mentioned materials and related configurations, the price will be different. The magnetic flap level gauge is made of a series of different materials such as 304 and 316L stainless steel, PP, PVC, stainless steel, PTFE lining, etc. It is suitable for different measuring media and working conditions (temperature, pressure, etc.). It needs to be equipped with a vacuum jacket or Steam jacket, the price will be adjusted accordinglyWorking principle of magnetic flap level gaugeWhen the liquid level in the container under test rises or falls, the magnetic float in the body tube of the level gauge will rise or fall accordingly. The permanent magnet steel in the float is transferred to the magnetic flap indicator through the magnetic coupling to drive the red and white When the liquid level rises, the tilting column turns from white to red. When the liquid level drops, the tipping column changes from red to white. The red and white junction of the indicator is the actual height of the liquid level in the container to achieve a clear indication of the liquid level. The axial distance between two adjacent indicator plates is 10mm, and the liquid level indication error is &; plusmn company; 10mm.According to the principle of buoyancy, the float moves up and down in the measuring tube as the liquid level rises and falls. The permanent magnet in the float drives the red and white column to rotate 180 degrees through magnetic coupling. When the liquid level rises, the column changes from white to red, and when it falls, the column changes from red to white to realize the liquid level indicationHow much is the magnetic flap level gauge? 2. Upper and lower limit switch output. The magnetic float moves with the liquid level to make the reed switch installed in the setting position of the liquid level gauge riser actuate to achieve switch control or alarm3. Remote transmission of level gaugeInstall the transmitter on the magnetic flap level gauge. The transmitter is composed of two parts: the sensor and the converter. It moves up and down by magnetic floats. Through magnetic coupling, the measuring elements in the catheter act in sequence to obtain the change of the resistance signal, which is converted into a 0-10 or 4-20mA standard current signal output, which is connected with a digital display instrument or a computer to realize remote display and control.
Post time: 21-09-21